Most of the volumetric apparatus available in the United States is calibrated at 20,although the temperatures generally prevailing in laboratories more nearly approach 25.To minimize volumetric error,the temperature should be the same for the volumetric apparatus,the material being prepared,the solvents being used to prepare the volumetric solutions,the area in which they are prepared,and the final volume adjustment.
Use To attain the degree of precision required in many Pharmacopeial assays involving volumetric measurements and directing that a quantity be accurately measured,the apparatus must be chosen and used with care.Aburet should be of such size that the titrant volume represents not less than 30%of the nominal volume.Where less than 10mLof titrant is to be measured,a 10-mLburet or a microburet generally is required.
The design of volumetric apparatus is an important factor in assuring accuracy.For example,the length of the graduated portions of graduated cylinders should be not less than five times the inside diameter,and the tips of burets and pipets should restrict the outflow rate to not more than 500µLper second.
Standards of Accuracy The capacity tolerances for volumetric flasks,transfer pipets,and burets are those accepted by the National Institute of Standards and Technology (Class A),*as indicated in the accompanying tables.Use Class Avolumetric apparatus unless otherwise specified in the individual monograph.For plastic volumetric apparatus the accepted capacity tolerances are Class B.
The capacity tolerances for measuring (i.e.,graduated)pipets of up to and including 10-mLcapacity are somewhat larger than those for the corresponding sizes of transfer pipets,namely,10,20,and 30µLfor the 2-,5-,and 10-mLsizes,respectively.
Transfer and measuring pipets calibrated to delivershould be drained in a vertical position and then touched against the wall of the receiving vessel to drain the tips.Volume readings on burets should be estimated to the nearest 0.01mLfor 25-and 50-mLburets,and to the nearest 0.005mLfor 5-and 10-mLburets.Pipets calibrated to containare called for in special cases,generally for measuring viscous fluids like syrups;however,a volumetric flask may be substituted for a to containpipet.In such cases,the pipet or flask should be washed clean,after draining,and the washings added to the measured portion.
* See Testing of Glass Volumetric Apparatus,N.B.S.Circ.602,April 1,1959,and NTIS COM-73-10504,National Technical Information Service.